by Dr. Tonny Mulder & Meike Mulder
ALSO: Read their descriptions carefully to get a feel of what every component does.
ALL COMPONENTS PRESENT?
1. Place the Powerbank in the EEG amplifier and connect by inserting the internal USB.
2. When using Optical cable: Insert one end of the Optical cable into the back of EEG amplifier and the other end into the adapter. Otherwise: skip this step.
3. Insert the LED side of the Serial cable into the EEG computer and the other end into the stimulus computer (your own computer).
4. Insert the cable of the Electrode box into the front of the EEG amplifier. Be careful when removing this cable later. Press in the sides before pulling!
5. Turn on the EEG amplifier with the switch under the lid. The light should turn red. (RED = ON)
6. Connect the EEG laptop to the EEG amplifier by going to the EEG laptops' Wi-Fi settings and selecting the name of the amplifier (see inside the amplifier for the name and password). When connected, it should say: "No internet, secured". The light in the amplifier should turn green (GREEN = connected).
NOTE: Make sure the EEG laptop does not automatically (re)connect to eduroam or another Wi-Fi network during the recording. Do this by turning off "Connect automatically" for all other Wi-Fi networks.
1. Powerbank in EEG amplifier.
2. Optional: Optical cable from adapter to EEG amplifier.
3. Serial cable from EEG (LED) to stimulus laptop.
6. Name and password EEG amplifier
4. Electrode box to EEG amplifier.
NOTE: Please dubbelcheck if the cables are inserted correctly and the EEG laptop is connected to the correct EEG amplifier!
NOTE 2: Make sure you understand this set-up before proceeding. For help, refer back to the explanations given in the Component Checklist!
ALL COMPONENTS CONNECTED CORRECTLY? UNDERSTAND THE SET-UP?
The different placements of the electrodes on the cap are based on the International "10-20 system" for EEG recording. For a detailed explanation, click here -->
Click figure to enlarge
You will have to decide where you want to measure. With a limited number of electrodes, you can't record all areas. So select your recording locations wisely.
Select literature that studied effects in a similar setting than yours and see where they placed the electrodes. Also, you can use the placements on which they found the strongest effect.
KEEP IN MIND: Be very precise on what kind of effect you are looking for: using visual stimuli does not necessarily mean the effect will be found over the visual cortex. For example, if you use words or pictures with different meanings in different context the effect will more likely take place over the temporal cortex. Also auditory ERPs are best recorded on PZ and are less effectively recorded on lateral positions. So: base this is on literature!
The best signal is often measured at other locations than directly over the brain area where it is generated. As it passes through the skull and scalp it can be smeared out. The result of this is that EEG has low spatial selectivity (meaning it is hard to determine where the measured signal comes from exactly). On the upside, this also means that, even if you place an electrode next to where the maximum effect actually is, you will still pick it up. Consequently, placing two electrodes right next to each other does not give you much more information. Keep this in mind as well.
This file is needed for the recorder software to know which channels you want to measure.
1. Download the Configuration-file from Canvas, copy this on a USB stick and transfer it to the EEG computer. There, save it on the desktop.
IMPORTANT: Create a DUPLICATE and RENAME this file under a UNIQUE name, preferably containing the name of your group (e.g. PsyPhyC1.xml). This will be the configuration-file you can edit.
2. Next, download and read the Psy Phy Config file (XML).docx from Canvas AND the points below to determine what information you can/must change in the configuration-file and what information to leave unchanged:
The green cap has a medium fit (56 cm) and fits most people. The light red cap is the largest (60 cm), and the dark red cap is the smallest (52 cm). The cap should be tight enough so the electrodes are close to the scalp but still comfortable for the subject to wear. Let the subject test out the caps.
After a cap is chosen, let the participant take off the cap, and insert the electrodes on the locations chosen by you. Then let the participant put on the cap with electrodes.
Measure the head size of the subject from nasion to inion, and from the left tip of the ear to the tip of the right ear. The location of the Cz electrode should be in the center of these measures. Move the cap around if necessary and make sure it fits tight and symmetrical.
Let the subject place the chin band.
Now you are ready to gel the electrodes.
1: Fill the curved syringe with electrode gel.
2: Place the tip of the syringe through the electrode on the skull. Ask the subject whether he/she feels the tip.
3: Apply the gel while retracting the syringe.
4: The reference and bias electrodes (red and blue earclips) are gelled by appying a drop of gel on the metal electrode disk and placing them on each of the earlobes.
1. Open the recording software by dragging your own configuration-file to the Fysiorecorder EEG software on the desktop.
2. Check if the right channels are going to be recorded.
3. Start a recording by pressing "Start Block" on the bottom left.
The recording has now started! The light on the amplifier should turn yellow (YELLOW = sending data). Give the signal a moment to stabilise. Be patient.
KEEP IN MIND: you can NOT pause the recording by pressing "Stop Block". This will automatically save the data and CLOSE the software! From this point on, you can leave the recording running until the participant has completed the experiment.
4. Check whether the signal quality is good enough.
4.1 Impedance Check:
4.2 Visibility of common artifacts/signals:
5. Once every channels shows a good signal, the participant can start the experiment on the stimulus laptop.
Once the participant has completed the whole experiment, the EEG recording can be stopped and saved:
6. To stop the recording, press "Stop Block" in the bottom left. This pop-up should appear. Press "OK".
When a recording is finished, always complete these steps right after:
1. First, remove the reference & bias, then remove the cap with the rest of the electrodes.
2. Remove as much gel from the hair of the participant as possible using paper tissue or a towel.
3. Gently rinse the electrodes clean with warm water. IMPORTANT: the electrodes themselves can get wet, however keep the plugs (which are plugged into the electrode box) DRY at all times!
4. Dry off the electrodes immediately after washing and double check if no gel remains on the electrode. When old gel dries up on the electrode, it can block the signal during the next recording.
5. Clean the syringe with water and dry it off with paper tissue.
6. Clean the cap by removing gel rests with a bit of tissue. If possible: when a lot of gel has gotten on the cap, wash it out with warm water by hand, and leave it out to dry before measuring the next participant.